HACH SC100 MANUAL PDF
View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Published (Last):||23 January 2014|
|PDF File Size:||2.10 Mb|
|ePub File Size:||17.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hex nut 4Cat. Index Index Numerics 90 Degree Detector Section 1 Specifications Section 1 Specifications Specifications are subject to change without notice.
Hach-Lange SC User Manual | 58 pages
Section 5 System Startup 5. The data logs store the measurement data at selected intervals. Tell us about it. Installation Figure 8 Panel Mounting the Controller 1 2 3 4 5 6 7 8 9 8 10 11 1. Page 12 Installation 3. Appendix A Modbus Protocol A.
Hach-Lange SC : Owner manual
The event log stores a variety of events that occur on the devices such as configuration changes, alarms, and warning conditions. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed ec100 a different value.
The time is available only in hour military format.
Supplied mounting hardware is shown in Figure 6, Figure 7, and Figure 8. Failed fuses are an indication that an equipment problem could exist. In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Ahch ensure that the protection provided by this equipment is not impaired, do not use or zc100 this equipment in any manner other than that specified in this manual.
Page 14 Installation 3. Manhal vertical or horizontal 5. After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system. Section 6 Replacement Parts 6.
Operates relays in response to the measured parameter. The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair.
Choose a number between 1 and as the address each source must have a different address then press ENTER. Section 4 Operation 4. Page 7 Section 1 Specifications Specifications hacb subject to change without notice. For the features manial for each individual digital network port, refer to the instruction sheet supplied with the selected network card.
Replacing The Lamp Assembly Page 6 Table of Contents Section 5 Maintenance Figure 22 Front of the Controller 1 sc 5 2 6 3 7 4 1. The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used.
Twist the bulb in a counterclockwise direction, pulling out slightly, until it is manul from the housing.
Hach-Lange SC 100 User Manual
Page 20 Jach 3. Section 4 Operation 4. Pull the lamp leads and connector through the hole in the lamp housing. Turbidimeter Installation Information Installation 3. Page 39 Operation 4. Under normal use, Hach recommends replacing the lamp once a year to maintain peak performance. Mounting Foot 2Cat.
In addition, the passcode also regulates function selection for relay options. Page 30 Operation 4. Fuses F1 and F2.